| ||Format||Pages||Price|| |
|3||$39.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$39.00||  ADD TO CART|
|Standard + Redline PDF Bundle||6||$46.80||  ADD TO CART|
Significance and Use
5.1 This test method is suitable for quality control, as a specification test, and for research. Acid numbers indicate the extent of any neutralization reaction of the polyol with acids. The results of this method measure batch-to-batch uniformity and are used as correction factors in calculating true hydroxyl numbers.
1.1 This test method measures the acidic constituents in polyether polyols and reports the results as acid number. The typical acid number range is 0-0.1 mg KOH/g sample. (See .)
1.2 The values stated in SI units are to be regarded as standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Note 1: There is no known ISO equivalent to this standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D883 Terminology Relating to Plastics
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
ICS Number Code 19.020 (Test conditions and procedures in general); 83.040.01 (Raw materials for rubber and plastics in general)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7253-16, Standard Test Method for Polyurethane Raw Materials: Determination of Acidity as Acid Number for Polyether Polyols, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top