| ||Format||Pages||Price|| |
|9||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||9||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||18||$52.80||  ADD TO CART|
Significance and Use
5.1 Exposure to beryllium can cause a potentially fatal disease, and occupational exposure limits for beryllium in air and on surfaces have been established to reduce exposure risks to potentially affected workers (. Sampling and analytical methods for beryllium are needed in order to meet the challenges relating to exposure assessment and risk reduction. Sampling and analysis methods, such as the procedure described in this test method, are desired in order to facilitate on-site and fixed-site laboratory measurement of trace beryllium. Beryllium analysis results can then be used as a basis for exposure assessment and protection of human health. , )
1.1 This test method is intended for use in the determination of beryllium by sampling workplace air and surface dust.
1.2 This test method assumes that air and surface samples are collected using appropriate and applicable ASTM International standard practices for sampling of workplace air and surface dust. These samples are typically collected using air filter sampling, vacuum sampling or wiping techniques. See Guide for guidance on air sampling strategies, and Guide for guidance on selection of surface sampling techniques.
1.3 Determination of beryllium in soil is not within the scope of this test method. See Test Method for determination of beryllium in soil samples.
1.4 This test method includes a procedure for extraction (dissolution) of beryllium in weakly acidic medium (pH of 1 % aqueous ammonium bifluoride is 4.8), followed by field analysis of aliquots of the extract solution using a beryllium-specific-optically fluorescent dye.
1.5 The procedure is suitable for on-site use in the field for occupational and environmental hygiene monitoring purposes. The method is also applicable for use in fixed-site laboratories.
1.6 No detailed operating instructions are provided because of differences among various makes and models of suitable fluorometric instruments. Instead, the analyst shall follow the instructions provided by the manufacturer of the particular instrument. This test method does not address comparative accuracy of different devices or the precision between instruments of the same make and model.
1.7 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.8 This test method contains notes that are explanatory and not part of mandatory requirements of the standard.
1.9 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D1356 Terminology Relating to Sampling and Analysis of Atmospheres
D4840 Guide for Sample Chain-of-Custody Procedures
D5337 Practice for Flow Rate Adjustment of Personal Sampling Pumps
D6966 Practice for Collection of Settled Dust Samples Using Wipe Sampling Methods for Subsequent Determination of Metals
D7035 Test Method for Determination of Metals and Metalloids in Airborne Particulate Matter by Inductively Coupled Plasma Atomic Emission Spectrometry (ICP-AES)
D7144 Practice for Collection of Surface Dust by Micro-vacuum Sampling for Subsequent Metals Determination
D7296 Practice for Collection of Settled Dust Samples Using Dry Wipe Sampling Methods for Subsequent Determination of Beryllium and Compounds
D7458 Test Method for Determination of Beryllium in Soil, Rock, Sediment, and Fly Ash Using Ammonium Bifluoride Extraction and Fluorescence Detection
D7659 Guide for Strategies for Surface Sampling of Metals and Metalloids for Worker Protection
D7707 Specification for Wipe Sampling Materials for Beryllium in Surface Dust
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
ICS Number Code 13.040.30 (Workplace atmospheres)
UNSPSC Code 12141501(Beryllium Be); 77121500(Air pollution)
ASTM D7202-15, Standard Test Method for Determination of Beryllium in the Workplace by Extraction and Optical Fluorescence Detection, ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top