| ||Format||Pages||Price|| |
|PDF Version||7||$42.00||  ADD TO CART|
|Print Version||7||$42.00||  ADD TO CART|
Significance and Use
4.1 This RIPT method assesses the potential of skin sensitization with a particular medical product by repeated topical applications to the skin of selected subjects. This is a procedure that has the potential to detect many, but not all, sensitzers. This requires multiple applications to induce a cell-mediated Type IV immune response sufficient to cause an allergic reaction.
4.2 In general, the sensitization procedure requires 10 multiple 48-h (72-h on weekends) applications of patches containing the study material over a three-week induction phase. Induction is followed by approximately a 21 day rest phase to allow the development of any latent sensitization. Study subjects are then challenged by the application of two consecutive 48-h patches of the study material to naive sites. Responses are evaluated and graded after the removal of each consecutive 48-h patch application.
1.3 During the performance of the Human Repeat Insult Patch Test (RIPT) for determining sensitization, investigators are confronted with skin responses that represent skin irritation (non-immunologic responses) or allergic contact dermatitis (ACD). The numerical scoring system for grading the intensity of both are similar and test facilities may vary in their scores that describe intensities of allergic and irritant skin responses. The hallmark of a mild allergic contact dermatitis is a sustained palpable erythematous reaction. Delayed-type allergic contact reactions from patch tests have intensity characteristics that favor scores of higher values for longer periods of time and typically do not produce a minimal score (score of 1, a just-perceptible erythema) for short durations (less than 48 h). It is the responsibility of the investigator to evaluate the scores in light of irritant reactions so that the responses are allergic in nature and not irritant. The investigator should denote a final score as either due to contact allergy or irritation. Paragraphs 9.5-9.5.5 describe a commonly used scoring system and discuss allergic and irritant responses in detail.
1.4 The Draize RIPT was published in 1944 as an attempt to decrease the frequency ACD.2 The test techniques at that time were just being validated and this experimental design was largely empiric.3 The principle of the test is as follows:
1.6 In 1945, Henderson and Riley4 demonstrated that a test panel sample size of 30 000 subjects would have to be employed to ensure statistically that there would be no more than 0.1 % sensitization. If there are no allergic responses in a test panel of 200 subjects with exposures comparable to those of the population, then there could be as many as 1.5 allergic reactions per 100 users.
1.7 All medical devices must be safe and effective for their intended use. Since medical devices such as gloves come in contact with human tissue, they should be tested for biocompatibility in animals first. The human repeat insult patch test (RIPT) is one test that can be used to test rubber gloves for skin sensitization to chemicals used in the manufacture of gloves.
ICS Number Code 11.140 (Hospital equipment)
UNSPSC Code 42132203(Medical exam or non surgical procedure gloves)