| ||Format||Pages||Price|| |
|5||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$45.00||  ADD TO CART|
Significance and Use
This test method is designed to determine the amount of residual powder and non-powder solids found on medical gloves.
This test method is suitable and designed as a reference method to evaluate samples of medical gloves.
The mass found using Procedure II, for powdered gloves, is assumed to be a combination of water-insoluble residue remaining after the manufacturing process, former release agents and donning powder.
1.1 This test method covers the determination of average powder or filter-retained mass found on a sample of medical gloves as described in the introduction.
1.2 The average powder mass per glove is reported in milligrams.
1.3 The safe and proper use of medical gloves is beyond the scope of this test method.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D4483 Practice for Evaluating Precision for Test Method Standards in the Rubber and Carbon Black Manufacturing Industries
ICS Number Code 11.140 (Hospital equipment)
UNSPSC Code 42132200(Medical gloves and accessories)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D6124-06(2011), Standard Test Method for Residual Powder on Medical Gloves, ASTM International, West Conshohocken, PA, 2011, www.astm.orgBack to Top