Active Standard ASTM D5984 | Developed by Subcommittee: D02.06
Book of Standards Volume: 05.02
Historical (view previous versions of standard)
Significance and Use
New and used petroleum products can contain basic constituents that are present as additives or as degradation products formed during service. The amount of these additives in an oil can be determined by titrating against an acid. The base number is a measure of the amount of basic substance in the oil, always under the conditions of the test. A decrease in base number is often used as a measure of lubricant degradation, but any condemning limits must be empirically established.
This test method uses reagents that are considered less hazardous than most reagents used in alternate base number methods. It uses pre-packaged reagents to facilitate base number determinations in the field where scientific equipment is unavailable and quick results are at a premium.
Note 1—Results obtained by this test method3 are similar to those obtained by Test Method D 2896
1.1 This test method covers a procedure for determining the basic constituents in petroleum products in the field or laboratory using a pre-packaged test kit. The test uses a micro-titration resulting in a visual end point facilitated by a color indicator.
1.1.1 This test method covers base numbers from 0 to 20. It can be extended to higher ranges by diluting the sample or by using a smaller sample size; however, the precision data were obtained for base numbers up to 20.
1.2 This test method can be used to indicate relative changes that occur in an oil during use under oxidizing conditions. Although the test is performed under closely specified conditions with standardized reagents, the test method does not measure an absolute basic property that can be used to predict performance of an oil under service conditions. No general relationship between bearing corrosion and base number is known.
1.3 The values stated in SI units are to be regarded as the standard. The values given in parentheses are for information only.
This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D2896 Test Method for Base Number of Petroleum Products by Potentiometric Perchloric Acid Titration
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
ICS Number Code 71.040.40 (Chemical analysis)