| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
Significance and Use
5.1 The pour point of a petroleum product is an index of the lowest temperature of its utility for certain applications. Flow characteristics, like pour point, can be critical for the correct operation of lubricating oil systems, fuel systems, and pipeline operations.
5.2 Petroleum blending operations require precise measurement of the pour point.
5.3 This test method can determine the pour point of the test specimen with a resolution of 1.0 °C.
5.4 Test results from this test method can be determined at either 1 °C or 3 °C intervals.
5.5 This test method yields a pour point in a format similar to Test Method D97/IP15 when the 3 °C interval results are reported.
5.6 This test method has better repeatability and reproducibility relative to Test Method D97/IP15 as measured in the 1998 interlaboratory test program. (See Section 13.)
1.1 This test method covers the determination of pour point of petroleum products by an automatic instrument that tilts the test jar during cooling and detects movement of the surface of the test specimen with an optical device.
1.2 This test method is designed to cover the range of temperatures from −66 °C to +51 °C; however, the range of temperatures included in the 1992 interlaboratory test program only covered the temperature range from −39 °C to +6 °C, and the range of temperatures included in the 1998 interlaboratory test program was −51 °C to −11 °C. (See Section 13.)
1.3 Test results from this test method can be determined at 1 °C or 3 °C intervals.
1.4 This test method is not intended for use with crude oils.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D97 Test Method for Pour Point of Petroleum Products
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
D6708 Practice for Statistical Assessment and Improvement of Expected Agreement Between Two Test Methods that Purport to Measure the Same Property of a Material
ICS Number Code 75.080 (Petroleum products in general)
UNSPSC Code 15101500(Petroleum and distillates)
ASTM D5950-14, Standard Test Method for Pour Point of Petroleum Products (Automatic Tilt Method), ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top