| ||Format||Pages||Price|| |
|11||$49.00||  ADD TO CART|
|Hardcopy (shipping and handling)||11||$49.00||  ADD TO CART|
Significance and Use
In order to be certain that the end user of analytical results obtained from using an ASTM Committee D-19 test method can be confident that the values have been obtained through a competent application of the test method, a demonstration of the proficiency of the analytical system must be performed. Appropriate proficiency is demonstrated by achievement of performance criteria derived from results of the test method collaborative study. The QC measures specified in this practice must be included in each ASTM test method, as applicable, to ensure the quality of measurements.
In order for users of D-19 test methods to achieve consistently valid results, a minimum level of QC must be performed. This minimum level of QC is stipulated in this practice and by the taskgroups developing D-19 test methods. If the specific requirements outlined in this practice are not applicable to the test method, alternative QC must be defined in the test method.
1.1 This practice provides specific, mandatory requirements for incorporating quality control (QC) procedures into all test methods under the jurisdiction of Committee D-19.
1.2 ASTM has adopted the following:
|Policy on implementation of requirements for a quality control section in standard test methods generated by Committee D-19 on Water.|
|NEW COLLABORATIVE TESTING|
|OLDER VALIDATED METHODS|
1.3 Required QC sections in all applicable test methods are intended to achieve two goals. First, users of Committee D-19 test methods will be able to demonstrate a minimum competency in the performance of these test methods by comparison with collaborative study data. Second, all users of test methods will be required to perform a minimum level of QC as part of proper implementation of these test methods to ensure ongoing competency.
1.4 This practice contains the primary requirements for QC of a specific test method. In many cases, it may be desirable to implement additional QC requirements to assure the desired quality of data.
1.5 The specific requirements in this practice may not be applicable to all test methods. These requirements may vary depending on the type of test method used as well as the analyte being determined and the sample matrix being analyzed. See Explanation 1 in Appendix X1.
1.5.1 If there are compelling reasons why any of the specific QC requirements listed in this practice are not applicable to a specific test method, these reasons must be documented in the QC section of the test method.
1.5.2 With the approval of Committee D-19 on the recommendation of the D-19 Results Advisor and the Technical Operations section of the Executive Subcommittee, a statement giving the compelling reasons why compliance with all or specific points of this practice cannot be achieved will meet the requirements of both ASTM and this practice.
1.5.3 Test Methods developed prior to the approval of this practice with a QC Section that meet the requirements of Specification D5789 are considered in compliance with this Practice.
1.6 This practice is for use with quantitative methods and may not be applicable to qualitative test methods.
1.7 Presently, this practice is applicable primarily to chemical test methods. It is intended that, in future revisions, the practice will be expanded to include other methods such as microbiological methods.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3648 Practices for the Measurement of Radioactivity
D3856 Guide for Management Systems in Laboratories Engaged in Analysis of Water
D4375 Practice for Basic Statistics in Committee D19 on Water
D5789 Practice for Writing Quality Control Specifications for Standard Test Methods for Organic Constituents
D5810 Guide for Spiking into Aqueous Samples
ICS Number Code 13.060.45 (Examination of water in general)
ASTM D5847-02(2012), Standard Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis, ASTM International, West Conshohocken, PA, 2012, www.astm.orgBack to Top