| ||Format||Pages||Price|| |
|5||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$44.00||  ADD TO CART|
Significance and Use
Air quality standards for ozone have been promulgated by government authorities to protect the health and welfare of the public. Though ozone itself is a toxic material, it is often complex organic compounds that cause the symptoms of smog such as tearing and burning eyes. However, ozone is the predominant oxidant and is much more easily monitored than organic species. Since ozone concentrations are also correlated with other photochemical oxidant levels, it is the substance that is specified in air quality standards and regulations.
1.1 This test method describes the sampling and continuous analysis of the ozone content of the atmosphere at concentrations of 20 to 2000 μg of ozone/m3 (10 ppb (v) to 1 ppm (v)).
1.2 This test method is limited in application by its sensitivity to interferences as described below. This test method is not suitable for personal sampling because of instrument size and sensitivity to vibration and ambient temperature.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Some specific precautionary statements are presented in Section 8.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1356 Terminology Relating to Sampling and Analysis of Atmospheres
D1357 Practice for Planning the Sampling of the Ambient Atmosphere
D1914 Practice for Conversion Units and Factors Relating to Sampling and Analysis of Atmospheres
D3249 Practice for General Ambient Air Analyzer Procedures
D3670 Guide for Determination of Precision and Bias of Methods of Committee D22
D5011 Practices for Calibration of Ozone Monitors Using Transfer Standards
D5110 Practice for Calibration of Ozone Monitors and Certification of Ozone Transfer Standards Using Ultraviolet Photometry
U.S. Environmental Protection Agency StandardsEPA-600/4-83-003 Performance Test Results and Comparative Data for Designated Reference and Equivalent Methods for Ozone (NTIS: PB83166686)
Code of Federal Regulations40-CFR-Part 53.20
ICS Number Code 13.040.20 (Ambient atmospheres)
UNSPSC Code 77121501(Air quality management)
ASTM D5149-02(2008), Standard Test Method for Ozone in the Atmosphere: Continuous Measurement by Ethylene Chemiluminescence, ASTM International, West Conshohocken, PA, 2008, www.astm.orgBack to Top