| ||Format||Pages||Price|| |
|5||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$54.00||  ADD TO CART|
Significance and Use
5.1 This test method is designed to obtain styrene purity on the basis of impurities normally present in styrene and may be used for final product inspections and process control.
5.2 This test method will detect the impurities shown in , non-aromatic hydrocarbons containing ten carbons or less, and others where specific impurity standards are available. Absolute purity cannot be determined if unknown impurities are present.
1.1 This test method covers the determination of the impurities typically found in styrene and uses those values to determine the purity of styrene by difference by gas chromatography. Typical impurities in commercial styrene are found in .
1.2 This test method is applicable to impurity concentrations in the range of 0.001 to 1.00 % and for styrene purities of 99 % or higher when using the internal standard procedure.
1.3 In determining the conformance of the test results using this method to applicable specifications, results shall be rounded off in accordance with the rounding-off method of Practice .
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For a specific hazard statement, see Section .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D3437 Practice for Sampling and Handling Liquid Cyclic Products
D4307 Practice for Preparation of Liquid Blends for Use as Analytical Standards
D4790 Terminology of Aromatic Hydrocarbons and Related Chemicals
D6809 Guide for Quality Control and Quality Assurance Procedures for Aromatic Hydrocarbons and Related Materials
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E355 Practice for Gas Chromatography Terms and Relationships
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E1510 Practice for Installing Fused Silica Open Tubular Capillary Columns in Gas Chromatographs
ICS Number Code 71.080.15 (Aromatic hydrocarbons)
UNSPSC Code 12352005(Aromatic or heterocyclic compounds)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D5135-16, Standard Test Method for Analysis of Styrene by Capillary Gas Chromatography, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top