| ||Format||Pages||Price|| |
|13||$49.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$49.00||  ADD TO CART|
|Standard + Redline PDF Bundle||26||$58.80||  ADD TO CART|
Significance and Use
4.1 This guide recommends the use of statistical x-charts to graphically monitor test data determined for the ASTM reference blacks for those test methods given in Section . All laboratories are encouraged to utilize statistical x-charts and ASTM reference blacks because this enables a comparison of testing precision within and between laboratories. The guide describes practices for the use of repeatability and reproducibility limits and x-charts.
4.2 In addition to the calibration of a test method by physicochemical means, a statistical method for achieving calibration of a test method is presented (that is, normalization).
4.3 Poor test precision can be the result of poor repeatability or poor reproducibility or both. Causes may include inadequate operator training, improperly maintained equipment or laboratory environment, variation in sample preparation or analysis techniques, the lack of calibration or standardization of instrumentation, worn-out apparatus, reagents that do not meet specifications, different sources of instrumentation or equipment, and material heterogeneity. The sum of all sources of testing error is unique for an individual laboratory.
4.4 Precision data for ASTM Reference Blacks are found in . These include standard reference blacks (SRB’s) Series 8, HT and INR Iodine Standards. The HT or INR Iodine standards are recommended for monitoring iodine testing.
Note 1: Preferred precision values are bolded in .
1.1 This guide covers procedures for using the ASTM Standard Reference Blacks (SRBs) and the HT and INR Iodine Number Standards to continuously monitor the precision of those carbon black test methods for which reference values have been established. It also offers guidelines for troubleshooting various test methods.
1.2 This guide establishes procedures for the use of x-charts to continuously monitor those tests listed in Section for within-lab precision (repeatability) and between-lab accuracy (reproducibility).
1.3 This guide provides a statistical procedure for improving test reproducibility when a laboratory cannot physically calibrate its apparatus to obtain the reference values of the ASTM reference blacks, within the ranges given in this guide.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1510 Test Method for Carbon Black--Iodine Adsorption Number
D1513 Test Method for Carbon Black, Pelleted--Pour Density
D1765 Classification System for Carbon Blacks Used in Rubber Products
D2414 Test Method for Carbon Black--Oil Absorption Number (OAN)
D3037 Test Method for Carbon Black--Surface Area by Nitrogen Adsorption
D3191 Test Methods for Carbon Black in SBR (Styrene-Butadiene Rubber)--Recipe and Evaluation Procedures
D3192 Test Methods for Carbon Black Evaluation in NR (Natural Rubber)
D3265 Test Method for Carbon Black--Tint Strength
D3493 Test Method for Carbon Black--Oil Absorption Number of Compressed Sample (COAN)
D3765 Test Method for Carbon Black--CTAB (Cetyltrimethylammonium Bromide) Surface Area
D4820 Test Methods for Carbon Black--Surface Area by Multipoint B.E.T. Nitrogen Adsorption
D5816 Test Methods for Carbon Black--External Surface Area by Multipoint Nitrogen Adsorption
D6556 Test Method for Carbon Black--Total and External Surface Area by Nitrogen Adsorption
ICS Number Code 83.040.20 (Rubber compounding ingredients)
UNSPSC Code 12171603(Carbon black)
ASTM D4821-14, Standard Guide for Carbon Black—Validation of Test Method Precision and Bias, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top