Developed by Subcommittee: D02.04.0H
WITHDRAWN, NO REPLACEMENT
Formerly under the jurisdiction of Committee D02 on Petroleum Products and Lubricants, this test method was withdrawn in January 2013 in accordance with section 10.5.3.1 of the Regulations Governing ASTM Technical Committees, which requires that standards shall be updated by the end of the eighth year since the last approval date.
This test method covers the use of gas chromatography to determine the amount of diesel fuel in used engine lubricating oil. This test is limited to SAE 30 oil. The diesel fuel diluent is analyzed at concentrations up to 12 mass???%.
1.1 This test method covers the use of gas chromatography to determine the amount of diesel fuel in used engine lubricating oil. This test is limited to SAE 30 oil. The diesel fuel diluent is analyzed at conventrations up to 12 mass %.
Note 1—This test method may be applicable to higher viscosity grade oils. However, such oils were not included in the program used to develop the precision statement.
1.2 This test method is limited to gas chromatographs equipped with flame ionization detectors and temperature programmable ovens.
Note 2—The use of other detectors and instrumentation has been reported. However, the precision statement applies only when the instrumentation specified is employed.
1.3 There is some overlap of the boiling ranges of diesel fuel and SAE 30 engine oils. Moreover, the boiling range of SAE 30 oils from various sources can vary appreciably. As a result, the calibration can be altered by as much as 2 %, in terms of fuel dilution. When testing unknown or mixed brands of used engine oil, it should be realized that the precision of the method may be poorer than the precision obtained when calibrating with a new oil representative of the used oil being tested.
1.4 The values stated in SI units are to be regarded as the standard. The values stated in inch-pound units are for information only.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D4626 Practice for Calculation of Gas Chromatographic Response Factors
E260 Practice for Packed Column Gas Chromatography
E355 Practice for Gas Chromatography Terms and Relationships
E594 Practice for Testing Flame Ionization Detectors Used in Gas or Supercritical Fluid Chromatography
E1510 Practice for Installing Fused Silica Open Tubular Capillary Columns in Gas Chromatographs