| ||Format||Pages||Price|| |
|6||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$43.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$51.60||  ADD TO CART|
Significance and Use
4.1 For the broad range of commercial rubber grade carbon blacks, tint strength is highly dependent upon particle size. Tint strength can be used as an indication of particle size; however, tint strength is also dependent on structure and aggregate size distribution. Therefore, differences in tint strength within grades of carbon black may reflect differences other than particle size.
Note 1: This test method was developed primarily for the characterization of N100, N200, and N300 series carbon blacks.
4.2 Tint strength values within the carbon black industry have been developed using a Hoover Muller apparatus which is used to prepare carbon black-zinc oxide pastes. A new mixing apparatus, SpeedMixer (DAC 150 FVZ), and a corresponding procedure have been extensively studied within D24 and shown to provide equivalent tint strength for all carcass or soft blacks and most tread blacks with the exception of higher surface area N100 types and specialty blacks. Therefore, it is the responsibility of the user of this alternate apparatus to ensure their products will adequately disperse. Disputes arising between a user and producer should be resolved using the Hoover Muller apparatus until ASTM develops adequate precision statements.
4.3 The term ITRB is used in the entire text for both, the original ITRB, used as the first reference material for tint testing, but which is now used up, and the successor reference material, ITRB2. Wherever required, like in calculations where it is crucial to differentiate between the original ITRB and ITRB2, this will be clearly mentioned in the text of the test procedure.
1.1 This test method covers the determination of the tint strength of carbon black relative to an industry tint reference black (ITRB).
1.2 The values stated in SI units are to be regarded as the standard. The values given in parentheses are for information only.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1799 Practice for Carbon Black--Sampling Packaged Shipments
D1900 Practice for Carbon Black--Sampling Bulk Shipments
D4483 Practice for Evaluating Precision for Test Method Standards in the Rubber and Carbon Black Manufacturing Industries
D4821 Guide for Carbon Black--Validation of Test Method Precision and Bias
ICS Number Code 83.040.20 (Rubber compounding ingredients)
UNSPSC Code 12171603(Carbon black)
ASTM D3265-14, Standard Test Method for Carbon Black—Tint Strength, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top