Active Standard ASTM D3217 / D3217M | Developed by Subcommittee: D13.58
Book of Standards Volume: 07.01
Historical (view previous versions of standard)
Significance and Use
5.1 Both the loop breaking tenacity and the knot breaking tenacity, calculated from the breaking force measured under the conditions specified herein and the linear density of the fiber, are fundamental properties that are used to establish limitations on fiber-processing and upon their end-use applications. Physical properties, such as brittleness, not well defined by tests for breaking force and elongation can be estimated from the ratio of breaking tenacity measured in loop or knot tests, or both, and the normal tenacity as measured by Test Method D3822.
5.2 This test method is not recommended for acceptance testing of commercial shipments in the absence of reliable information on between-laboratory precision (see Note 3). In some cases the purchaser and the supplier may have to test a commercial shipment of one or more specific materials by the best available method, even though the method has not been recommended for acceptance testing of commercial shipments. In such a case, if there is a disagreement arising from differences in values reported by the purchaser and the supplier when using this test method for acceptance testing, the statistical bias, if any, between the laboratory of the purchaser and the laboratory of the supplier should be determined with each comparison being based on testing specimens randomly drawn from one sample of material of the type being evaluated.
1.1 These test methods cover the measurement of the breaking tenacity of manufactured textile fibers taken from filament yarns, staple, or tow fiber, either crimped or uncrimped, and tested in either a double loop or as a strand formed into a single overhand knot.
1.4 The basic distinction between the procedures described in these test methods and those included in Test Methods D2101 is the configuration of the specimen, that is, either as a double loop or in the configuration of a single overhand knot.
1.5 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D76 Specification for Tensile Testing Machines for Textiles
D123 Terminology Relating to Textiles
D1577 Test Methods for Linear Density of Textile Fibers
D1776 Practice for Conditioning and Testing Textiles
D2101 Test Method for Tensile Properties of Single Man-Made Textile Fibers Taken From Yarns and Tows
D2258 Practice for Sampling Yarn for Testing
D3333 Practice for Sampling Manufactured Staple Fibers, Sliver, or Tow for Testing
D3822 Test Method for Tensile Properties of Single Textile Fibers
D4849 Terminology Related to Yarns and Fibers
ICS Number Code 59.060.20 (Man-made fibres)
UNSPSC Code 11151500(Fibers); 41114632(Tensile strength tester)