| ||Format||Pages||Price|| |
|14||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||14||$50.00||  ADD TO CART|
|Standard + Redline PDF Bundle||28||$60.00||  ADD TO CART|
Significance and Use
5.1 This test method provides for the measuring of the minimum concentration of oxygen in a flowing mixture of oxygen and nitrogen that will just support flaming combustion of plastics. Correlation with burning characteristics under actual use conditions is not implied.
5.2 In this test method, the specimens are subjected to one or more specific sets of laboratory test conditions. If different test conditions are substituted or the end-use conditions are changed, it is not always possible by or from this test to predict changes in the fire-test-response characteristics measured. Therefore, the results are valid only for the fire-test-exposure conditions described in this test method.
1.1 This fire-test-response standard describes a procedure for measuring the minimum concentration of oxygen, expressed as percent volume, that will just support flaming combustion in a flowing mixture of oxygen and nitrogen.
1.2 This test method provides three testing procedures. Procedure A involves top surface ignition, Procedure B involves propagating ignition, and Procedure C is a short procedure involving the comparison with a specified minimum value of the oxygen index.
1.3 Test specimens used for this test method are prepared into one of six types of specimens (see Table 1).
1.4 This test method provides for testing materials that are structurally self-supporting in the form of vertical bars or sheet up to 10.5-mm thick. Such materials are solid, laminated or cellular materials characterized by an apparent density greater than 15 kg/m3.
1.5 This test method also provides for testing flexible sheet or film materials, while supported vertically.
1.6 This test method is also suitable, in some cases, for cellular materials having an apparent density of less than 15 kg/m3.
1.7 This test method measures and describes the response of materials, products, or assemblies to heat and flame under controlled conditions, but does not by itself incorporate all factors required for fire hazard or fire risk assessment of the materials, products, or assemblies under actual fire conditions.
1.8 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.9 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Specific hazards statement are given in Section 10.
1.10 Fire testing is inherently hazardous. Adequate safeguards for personnel and property shall be employed in conducting these tests.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D618 Practice for Conditioning Plastics for Testing
D1071 Test Methods for Volumetric Measurement of Gaseous Fuel Samples
D1622 Test Method for Apparent Density of Rigid Cellular Plastics
D4802 Specification for Poly(Methyl Methacrylate) Acrylic Plastic Sheet
E176 Terminology of Fire Standards
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
ISO StandardsISO 13943 Fire Safety--Vocabulary ISO 4589-2 Plastics--Determination of Flammability by Oxygen Index--Part 2, Ambient Temperatures ISO 7823-1 Poly(Methylmethacrylate) Sheets--Types, Dimensions and Characteristics--Part 1--Cast Sheets
ICS Number Code 83.080.01 (Plastics in general)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D2863-13, Standard Test Method for Measuring the Minimum Oxygen Concentration to Support Candle-Like Combustion of Plastics (Oxygen Index), ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top