| ||Format||Pages||Price|| |
|4||$38.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$38.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$45.60||  ADD TO CART|
Significance and Use
5.1 Test Method D2524 for testing wool fibers for tenacity is considered satisfactory for acceptance testing when the participating laboratories, using a reference wool, have shown acceptable between-laboratory precision. It is recommended that any program of acceptance testing be preceded by an interlaboratory check in the laboratory of the purchaser and the laboratory of the seller on replicate specimens of samples of the material to be evaluated. In cases of dispute, the statistical bias, if any, between the laboratory of the purchaser and the seller should be determined with each comparison being based on testing randomized specimens from one sample of material of the type being evaluated.
5.3 The basic differences between the procedures described in Test Method D2524 and those described in Test Method D1294 lie in the manner of clamping the bundles and the shorter gage length employed. The special clamps used in this method allow quicker and easier bundle preparation; however, for Test Method D1294 no special clamps are required. Closer agreement with single fiber tenacity is also obtained with Test Method D2524 than when using the procedure in Test Method D1294.
5.4 As the observed tenacity of fibers depends in part on the type of tensile testing machine used and the time required to break the specimen, results obtained with the different types of machines which may be used in this method will not necessarily agree. The machines specifically designed for bundle testing are CRL testers which operate at a loading rate of 1 kgf/s and therefore reach the breaking force at variable times in the order of 5 s. CRE and CRT type machines would be expected to produce somewhat different results not only because of the inherent difference in operation but because CRE and CRT type machines are to be operated at a rate to achieve the breaking load in 20 s.
1.2 This test method is especially adapted to the fiber bundle clamps and strength testing instruments specified, but may be used on other tensile testing machines when equipped with appropriate adapters to accommodate the prescribed clamps.
1.4 The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D123 Terminology Relating to Textiles
D540 Methods of Testing Man-Made Staple Fibers; Replaced by D 3660
D1294 Test Method for Tensile Strength and Breaking Tenacity of Wool Fiber Bundles 1-in. (25.4-mm) Gage Length
D1445 Test Method for Breaking Strength and Elongation of Cotton Fibers (Flat Bundle Method)
D1776 Practice for Conditioning and Testing Textiles
D2525 Practice for Sampling Wool for Moisture
D4845 Terminology Relating to Wool
ICS Number Code 59.060.10 (Natural fibres)
UNSPSC Code 11151508(Wool fibers)
ASTM D2524-13, Standard Test Method for Breaking Tenacity of Wool Fibers, Flat Bundle Method—