| ||Format||Pages||Price|| |
|6||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$54.00||  ADD TO CART|
Significance and Use
The determination of hydrocarbon impurities contained in toluene and mixed xylenes used as chemical intermediates and solvents is typically required. This test is suitable for setting specifications and for use as an internal quality control tool where aromatic monocyclic hydrocarbons are produced or are used. This test method is applicable for determining the impurities from the aromatic hydrocarbon production process. Typical impurities are alkanes containing 1 to 10 carbon atoms, benzene, toluene, ethylbenzene (EB), xylenes, and aromatic hydrocarbons containing nine carbon atoms.
Purity is commonly reported by subtracting the determined expected impurities from 100.00. However, a gas chromatographic analysis cannot determine absolute purity if unknown or undetected components are contained within the material being examined.
|Stationary phase||crosslinked polyethylene glycolA|
|Film thickness, μ||0.25|
|Diameter, mm||0.32 ID|
|Time 1, min||10|
|Time 2, min||10|
|Flow rate, mL/min||1.0|
|Sample size, μL||1.0|
|Analysis time, min||38|
|Linear velocity at 145°C, cm/s||20|
A Polyethylene glycol such as Carbowax 20 M available from most chromatographic suppliers, has been found suitable for this purpose.
1.1 This test method covers the determination of the total nonaromatic hydrocarbons, and trace monocyclic aromatic hydrocarbons in the purity of toluene and mixed xylenes by gas chromatography.
1.2 Nonaromatic aliphatic hydrocarbons containing 1 through 10 carbon atoms (methane through decanes) can be detected by this test method at concentrations ranging from 0.001 % to 2.500 weight %.
1.2.1 A small amount of benzene in mixed xylenes may not be distinguished from the nonaromatics and the concentrations are determined as a composite.
1.3 Monocyclic aromatic hydrocarbon impurities containing 6 through 9 carbon atoms (benzene through C9 aromatics) can be detected by this test method at individual concentrations ranging from 0.001 % to 1.000 weight %.
1.4 In determining the conformance of the test results using this method to applicable specifications, results shall be rounded off in accordance with the rounding-off method of Practice E29.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific hazard statement, see Section 9.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D841 Specification for Nitration Grade Toluene
D3437 Practice for Sampling and Handling Liquid Cyclic Products
D3797 Test Method for Analysis of o-Xylene by Gas Chromatography
D4492 Test Method for Analysis of Benzene by Gas Chromatography
D4790 Terminology of Aromatic Hydrocarbons and Related Chemicals
D5211 Specification for Xylenes for p-Xylene Feedstock
D6563 Test Method for Benzene, Toluene, Xylene (BTX) Concentrates Analysis by Gas Chromatography
D6809 Guide for Quality Control and Quality Assurance Procedures for Aromatic Hydrocarbons and Related Materials
D7504 Test Method for Trace Impurities in Monocyclic Aromatic Hydrocarbons by Gas Chromatography and Effective Carbon Number
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E260 Practice for Packed Column Gas Chromatography
E355 Practice for Gas Chromatography Terms and Relationships
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E1510 Practice for Installing Fused Silica Open Tubular Capillary Columns in Gas Chromatographs
ICS Number Code 71.080.15 (Aromatic hydrocarbons)
UNSPSC Code 12352005(Aromatic or heterocyclic compounds); 41115703(Gas chromatographs)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D2360-11, Standard Test Method for Trace Impurities in Monocyclic Aromatic Hydrocarbons by Gas Chromatography, ASTM International, West Conshohocken, PA, 2011, www.astm.orgBack to Top