| ||Format||Pages||Price|| |
|8||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||8||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||16||$54.00||  ADD TO CART|
Significance and Use
5.1 Because there are surface tension or kinematic viscosity differences, or both, between the primary standard ( ) and kinematic viscosity standards ( ), special procedures using master viscometers are required to “step-up” from the kinematic viscosity of the primary standard to the kinematic viscosities of oil standards.
5.2 Using master viscometers calibrated according to this practice, an operator can calibrate kinematic viscometers in accordance with Specifications .
5.3 Using viscosity oil standards established in this practice, an operator can calibrate kinematic viscometers in accordance with Specifications .
1.1 This practice covers the calibration of master viscometers and viscosity oil standards, both of which may be used to calibrate routine viscometers as described in Test Method and Specifications over the temperature range from 15 °C to 100 °C.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.2.1 The SI-based units for calibration constants and kinematic viscosities are mm2/s2 and mm 2/s, respectively.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific warning statements, see Section .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D445 Test Method for Kinematic Viscosity of Transparent and Opaque Liquids (and Calculation of Dynamic Viscosity)
D446 Specifications and Operating Instructions for Glass Capillary Kinematic Viscometers
D1193 Specification for Reagent Water
D1250 Guide for Use of the Petroleum Measurement Tables
D1480 Test Method for Density and Relative Density (Specific Gravity) of Viscous Materials by Bingham Pycnometer
D1590 Test Method for Surface Tension of Water
E1 Specification for ASTM Liquid-in-Glass Thermometers
E563 Practice for Preparation and Use of an Ice-Point Bath as a Reference Temperature
E644 Test Methods for Testing Industrial Resistance Thermometers
E1137 Specification for Industrial Platinum Resistance Thermometers
E1750 Guide for Use of Water Triple Point Cells
E2593 Guide for Accuracy Verification of Industrial Platinum Resistance Thermometers
E2877 Guide for Digital Contact Thermometers
ICS Number Code 17.060 (Measurement of volume, mass, density, viscosity)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D2162-14, Standard Practice for Basic Calibration of Master Viscometers and Viscosity Oil Standards, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top