WITHDRAWN, NO REPLACEMENT
| ||Format||Pages||Price|| |
|5||$54.00||  ADD TO CART|
This test method describes the determination of the gas content of electrical insulating liquids with a viscosity of 216 cSt or less at 100°C. Any gas that is nonreactive with a strong caustic solution may be determined.
Formerly under the jurisdiction of Committee D27 on Electrical Insulating Liquids and Gases and Subcommittee D27.03 on Physical Tests, this practice was withdrawn in May 2009 in accordance with section 10.5.3.1 of the Regulations Governing ASTM Technical Committees, which requires that standards shall be updated by the end of the eighth year since the last approval date.
1.1 This test method describes the determination of the gas content of electrical insulating liquids with a viscosity of 216 cSt or less at 100°C. Any gas that is nonreactive with a strong caustic solution may be determined.
Note 1—The test method has a bias for samples containing gases other than oxygen and nitrogen in atmospheric ratios due to differential solubility effects. Gases which react with KOH such as carbon dioxide will not be measured. Unsaturated hydrocarbons such as acetylene, if present, will react with KOH to a small degree and will result in an underestimation of the total gas present.
1.2 Warning—Mercury has been designated by EPA and many state agencies as a hazardous material that can cause central nervous system, kidney, and liver damage. Mercury, or its vapor, may be hazardous to health and corrosive to materials. Caution should be taken when handling mercury and mercury-containing products. See the applicable product Material Safety Data Sheet (MSDS) for details and EPA’s website (http://www.epa.gov/mercury/faq.htm) for additional information. Users should be aware that selling mercury or mercury-containing products, or both, in your state may be prohibited by state law.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D831 Test Method for Gas Content of Cable and Capacitor Oils
D923 Practices for Sampling Electrical Insulating Liquids
D1193 Specification for Reagent Water
D3613 Practice for Sampling Insulating Liquids for Gas Analysis and Determination of Water Content
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D1827-92(2002)e1, Standard Test Method for Gas Content (Nonacidic) of Insulating Liquids by Displacement with Carbon Dioxide (Withdrawn 2009), ASTM International, West Conshohocken, PA, 1992, www.astm.orgBack to Top