| ||Format||Pages||Price|| |
|5||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$54.00||  ADD TO CART|
Significance and Use
5.1 Uranium and plutonium are used in nuclear reactor fuel and must be analyzed to ensure that they meet acceptance criteria for isotopic composition as described in Specifications and . The criteria are set by mutual agreement between the manufacturer and end user (or between buyer and seller). This standard practice is used to separate chemically the isobaric interferences from 238U and 238Pu and from 241Am and 241Pu, and from other impurities prior to isotopic abundance determination by TIMS.
5.2 In facilities where perchloric acid use is authorized, the separation in Test Method may be used prior to isotopic abundance determination. Uranium and plutonium content as well as isotopic abundances using TIMS can be determined by using this separation practice and by following Test Methods , , or .
1.1 This practice is an alternative to Practice for the ion exchange separation in small mass samples (~5 μg of plutonium and up to 0.5 mg of uranium in 1 mL of solution) of uranium and plutonium from each other and from other impurities for subsequent isotopic abundance and content analysis by thermal ionization mass spectrometry (TIMS). In addition to being adapted to smaller sample sizes, this practice also avoids the use of hydrochloric acid (HCl) and hydrofluoric acid (HF) and does not require the use of two anion exchange columns as required in Practice .
1.2 In chemically unseparated samples isobaric nuclides at mass 238 (238U and 238Pu), and mass 241 (241Pu and 241Am) will be measured together thus compromising the accuracy of the results of isotopic composition of Pu. Therefore, chemical separation of elements is essential prior to isotopic analyses. Concentrations and volumes given in the paragraphs below can be modified for larger sample sizes, different types of anion exchange resin, etc.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C698 Test Methods for Chemical, Mass Spectrometric, and Spectrochemical Analysis of Nuclear-Grade Mixed Oxides ((U, Pu)O2)
C833 Specification for Sintered (Uranium-Plutonium) Dioxide Pellets
C859 Terminology Relating to Nuclear Materials
C1008 Specification for Sintered (Uranium-Plutonium) DioxidePellets--Fast Reactor Fuel
C1168 Practice for Preparation and Dissolution of Plutonium Materials for Analysis
C1347 Practice for Preparation and Dissolution of Uranium Materials for Analysis
C1411 Practice for The Ion Exchange Separation of Uranium and Plutonium Prior to Isotopic Analysis
C1415 Test Method for 238Pu Isotopic Abundance By Alpha Spectrometry
C1625 Test Method for Uranium and Plutonium Concentrations and Isotopic Abundances by Thermal Ionization Mass Spectrometry
C1672 Test Method for Determination of Uranium or Plutonium Isotopic Composition or Concentration by the Total Evaporation Method Using a Thermal Ionization Mass Spectrometer
D1193 Specification for Reagent Water
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1816-16, Standard Practice for The Ion Exchange Separation of Small Volume Samples Containing Uranium, Americium, and Plutonium Prior to Isotopic Abundance and Content Analysis, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top