| ||Format||Pages||Price|| |
|15||$49.00||  ADD TO CART|
|Hardcopy (shipping and handling)||15||$49.00||  ADD TO CART|
|Standard + Redline PDF Bundle||30||$58.80||  ADD TO CART|
Significance and Use
4.1 Measurement—The refractive index at any wavelength of a piece of homogeneous glass is a function, primarily, of its composition, and secondarily, of its state of annealing. The index of a glass can be altered over a range of up to 1×10-4 (that is, 1 in the fourth decimal place) by the changing of an annealing schedule. This is a critical consideration for optical glasses, that is, glasses intended for use in high performance optical instruments where the required value of an index can be as exact as 1×10-6. Compensation for minor variations of composition are made by controlled rates of annealing for such optical glasses; therefore, the ability to measure index to six decimal places can be a necessity; however, for most commercial and experimental glasses, standard annealing schedules appropriate to each are used to limit internal stress and less rigorous methods of test for refractive index are usually adequate. The refractive indices of glass ophthalmic lens pressings are held to 5×10-4 because the tools used for generating the figures of ophthalmic lenses are made to produce curvatures that are related to specific indices of refraction of the lens materials.
4.2 Dispersion—Dispersion-values aid optical designers in their selection of glasses (Note 1). Each relative partial dispersion-number is calculated for a particular set of three wavelengths, and several such numbers, representing different parts of the spectrum might be used when designing more complex optical systems. For most glasses, dispersion increases with increasing refractive index. For the purposes of this standard, it is sufficient to describe only two reciprocal relative partial dispersions that are commonly used for characterizing glasses. The longest established practice has been to cite the Abbe-number (or Abbe ν-value), calculated by:
where vD is defined in 3.2 and nD, nF, and nC are the indices of refraction at the emission lines defined in 3.2.
4.2.1 Some modern usage specifies the use of the mercury e-line, and the cadmium C′and F′ lines. These three lines are obtained with a single spectral lamp.
where ve is defined in 3.2 and ne, nF′, and nC′ are the indices of refraction at the emission lines defined in 3.2.
4.2.2 A consequence of the defining equations (Eq 1 and 2) is that smaller ν-values correspond to larger dispersions. For ν-values accurate to 1 to 4 %, index measurements must be accurate to 1×10-4; therefore, citing ν-values from less accurate test methods might not be useful.
1.1 This guide identifies and describes seven test methods for measuring the index of refraction of glass, with comments relevant to their uses such that an appropriate choice of method can be made. Four additional methods are mentioned by name, and brief descriptive information is given in Annex A1. The choice of a test method will depend upon the accuracy required, the nature of the test specimen that can be provided, the instrumentation available, and (perhaps) the time required for, or the cost of, the analysis. Refractive index is a function of the wavelength of light; therefore, its measurement is made with narrow-bandwidth light. Dispersion is the physical phenomenon of the variation of refractive index with wavelength. The nature of the test-specimen refers to its size, form, and quality of finish, as described in each of the methods herein. The test methods described are mostly for the visible range of wavelengths (approximately 400 to 780μm); however, some methods can be extended to the ultraviolet and near infrared, using radiation detectors other than the human eye.
22.214.171.124 Metricon2 system,
1.1.2 Test methods presented by name only (see Annex A1):
1.2 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.3 Warning—Refractive index liquids are used in several of the following test methods. Cleaning with organic liquid solvents also is specified. Degrees of hazard associated with the use of these materials vary with the chemical nature, volatility, and quantity used. See manufacturer's literature and general information on hazardous chemicals.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E167 Practice for Goniophotometry of Objects and Materials
E456 Terminology Relating to Quality and Statistics
ICS Number Code 81.040.30 (Glass products)
ASTM C1648-12, Standard Guide for Choosing a Method for Determining the Index of Refraction and Dispersion of Glass, ASTM International, West Conshohocken, PA, 2012, www.astm.orgBack to Top