| ||Format||Pages||Price|| |
|10||$43.00||  ADD TO CART|
|Hardcopy (shipping and handling)||10||$43.00||  ADD TO CART|
Significance and Use
This test method allows direct determination of the proportion of some individual phases in cement or portland-cement clinker. Thus it provides an alternative to the indirect estimation of phase proportion using the equations in Specification C150 (Annex A1).
This test method assumes that the operator is qualified to operate an X-ray diffractometer and to interpret X-ray diffraction spectra.
This test method may be used as part of a quality control program in cement manufacturing.
This test method may be used in predicting properties and performance of hydrated cement and concrete that are a function of phase composition.
QXRD provides a bulk analysis (that is, the weighted average composition of several grams of material). Therefore, results may not agree precisely with results of microscopical methods.
1.1 This test method covers direct determination of the proportion by mass of individual phases in portland cement or portland-cement clinker using quantitative X-ray (QXRD) analysis. The following phases are covered by this standard: alite (tricalcium silicate), belite (dicalcium silicate), aluminate (tricalcium aluminate), ferrite (tetracalcium aluminoferrite), periclase (magnesium oxide), gypsum (calcium sulfate dihydrate), bassanite (calcium sulfate hemihydrate), anhydrite (calcium sulfate), and calcite (calcium carbonate).
1.2 This test method specifies certain general aspects of the analytical procedure, but does not specify detailed aspects. Recommended procedures are described, but not specified. Regardless of the procedure selected, the user shall demonstrate by analysis of certified reference materials (CRM's) that the particular analytical procedure selected for this purpose qualifies (that is, provides acceptable precision and bias) (see Note 1). The recommended procedures are ones used in the round-robin analyses to determine the precision levels of this test method.
Note 1—A similar approach was used in the performance requirements for alternative methods for chemical analysis in Test Methods C114.
1.3 The values stated in SI units shall be regarded as the standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific hazards, see Section 9.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C114 Test Methods for Chemical Analysis of Hydraulic Cement
C150 Specification for Portland Cement
C183 Practice for Sampling and the Amount of Testing of Hydraulic Cement
C219 Terminology Relating to Hydraulic Cement
C670 Practice for Preparing Precision and Bias Statements for Test Methods for Construction Materials
E29 Practice for Using Significant Digits in Test Data to Determine Conformance with Specifications
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
ICS Number Code 91.100.10 (Cement. Gypsum. Lime. Mortar)
UNSPSC Code 30111601(Cement)
ASTM C1365-06(2011), Standard Test Method for Determination of the Proportion of Phases in Portland Cement and Portland-Cement Clinker Using X-Ray Powder Diffraction Analysis, ASTM International, West Conshohocken, PA, 2011, www.astm.orgBack to Top