WITHDRAWN, NO REPLACEMENT
| ||Format||Pages||Price|| |
|8||$54.00||  ADD TO CART|
This guide is one of a series on the application and evaluation of special nuclear material (SNM) monitors.
Formerly under the jurisdiction of Committee C26 on Nuclear Fuel Cycle, this practice was withdrawn in January 2014 in accordance with section 10.6.3.1 of the Regulations Governing ASTM Technical Committees, which requires that standards shall be updated by the end of the eighth year since the last approval date.
1.1 This guide is one of a series on the application and evaluation of special nuclear material (SNM) monitors. Other guides in the series are listed in Section 2, and the relationship of in-plant performance evaluation to other procedures described in the series is illustrated in Fig. 1. Hand-held SNM monitors are described in of Guide C1112, and performance criteria illustrating their capabilities can be found in Appendix X1.
1.2 The purpose of this guide to in-plant performance evaluation is to provide a comparatively rapid procedure to verify that a hand-held SNM monitor performs as expected for detecting SNM or alternative test sources or to disclose the need for repair. The procedure can be used as a routine operational evaluation or it can be used to verify performance after a monitor is calibrated.
1.3 In-plant performance evaluations are more comprehensive than daily functional tests. They take place less often, at intervals ranging from weekly to once every three months, and derive their result from multiple trials.
1.4 Note that the performance of both the hand-held monitor and its operator are important for effective monitoring. Operator training is discussed in Appendix X2.
1.5 The values stated in SI units are to be regarded as standard.
1.6 This standard does not purport to address all of the safety problems, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Note—The procedures shown “above” the user provide the user with information before acquiring a monitor, and those “below” assist the user to obtain continuing acceptable performance from the monitor.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C859 Terminology Relating to Nuclear Materials
C1112 Guide for Application of Radiation Monitors to the Control and Physical Security of Special Nuclear Material
C1189 Guide to Procedures for Calibrating Automatic Pedestrian SNM Monitors
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1237-99(2005), Standard Guide to In-Plant Performance Evaluation of Hand-Held SNM Monitors (Withdrawn 2014), ASTM International, West Conshohocken, PA, 2005, www.astm.orgBack to Top