Active Standard ASTM B555 | Developed by Subcommittee: B08.10
Book of Standards Volume: 02.05
Historical (view previous versions of standard)
Significance and Use
4.1 The thickness of a metal coating is often critical to its performance.
4.2 This procedure is useful for an approximate determination when the best possible accuracy is not required. For more reliable determinations, the following methods are available: Test Methods B487, B499, B504, and B568.
4.3 This test assumes that the rate of dissolution of the coating by the corrosive reagent under the specified conditions is always the same.
1.1 This guide covers the use of the dropping test to measure the thickness of electrodeposited zinc, cadmium, copper, and tin coatings.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
B487 Test Method for Measurement of Metal and Oxide Coating Thickness by Microscopical Examination of Cross Section
B499 Test Method for Measurement of Coating Thicknesses by the Magnetic Method: Nonmagnetic Coatings on Magnetic Basis Metals
B504 Test Method for Measurement of Thickness of Metallic Coatings by the Coulometric Method
B568 Test Method for Measurement of Coating Thickness by X-Ray Spectrometry
D1193 Specification for Reagent Water
ICS Number Code 25.220.40 (Metallic coatings)