Published: Jan 2010
| ||Format||Pages||Price|| |
|PDF Version (964K)||10||$25||  ADD TO CART|
|Complete Source PDF (41M)||10||$55||  ADD TO CART|
The use of intravascular stents in the femoropopliteal artery (FPA) continues to be controversial due to the potential fractures in the dynamic environment. The purpose of this study was to (1) develop a representative in-vitro model that simulates physiological motion of the FPA during knee and hip flexion and (2) use the model to characterize the types and ranges of stent distortion produced by extremity movement. This model eliminates inconsistencies often observed in cadaveric models and clinical subjects due to individual anatomical differences, and allows testing with large sample sizes in a controlled environment for variable (tubing length, material, diameter, and thickness) modification. A comparative evaluation of axial mechanical property and elasticity was conducted between the tubing intended to simulate arteries and the ex-vivo porcine carotid arteries, favoring the selection of silicone tubing. The model was assessed for its unstented and stented arterial bending and axial compression under three physiological motions: straight leg, walking (knee/hip flexion 70°/20°), and sitting/stair climbing (knee/hip flexion 90°/90°). Self-expanding nitinol stents implanted in the simulated mid-superficial femoral artery and popliteal artery (PA) of the model exhibit axial compression of 4.5±0.3 % and 7±0.3 % (knee/hip flexion 70°/20°), and 8.4±0.7 % and 8±0.2 % (knee/hip flexion 90°/90°). Stents implanted in the simulated PA exhibit bending of 40° and 74° from knee/hip angle changes to 70°/20° and to 90°/90°, respectively. The model demonstrated stent bending and compression as previously observed in cadaver studies. Additional analysis of stent motion (torsion, localized bending, radial compression) may be evaluated with more advanced imaging techniques and additional model development. The data generated in these analyses could support appropriate modes and parameters for stent fatigue testing, and better understanding of vascular device performance in the dynamic FPA.
stents, in-vitro model, superficial femoral artery, popliteal artery, biomechanical forces
Ph.D., MD, Abbott Vascular, Santa Clara, CA
Mach, Albert J.
BS, Abbott Vascular, Santa Clara, CA
MSE, Abbott Vascular, Santa Clara, CA
Ph.D., Abbott Vascular, Santa Clara, CA
Schwartz, Lewis B.
MD, Abbott Vascular, Santa Clara, CA
Paper ID: STP48985S