Published: Jan 2000
| ||Format||Pages||Price|| |
|PDF ()||17||$25||  ADD TO CART|
|Complete Source PDF (8.5M)||17||$136||  ADD TO CART|
The application of optical encoders in Charpy impact energy measurement has improved significantly the accuracy of “dial energy” determination. Instrumented strikers offer an alternative method of energy measurement which is accurate and reproducible for both conventional and miniature specimen testing while providing additional useful information such as general yield load, peak load, brittle fracture load, and brittle fracture arrest load. It has been observed that the total absorbed energy measured using these two technologies, while generally in good agreement, sometimes differs by a significant amount. The instrumented striker total absorbed energy has been found to be higher or lower than the optical encoder energy depending on the ductility of the test specimen and other factors. This paper examines and provides explanations for these energy differences. A summary of mechanisms for pendulum energy loss, other than in fracturing the test specimen, is provided along with estimates of the amount of energy which may be associated with each mechanism.
impact testing, instrumented striker, absorbed energy, optical encoder
President, MPM Technologies, Inc., State College, PA
Owner, Computational Mechanics, Inc., Julian, PA