This practice describes three methods for determining the relative spectral correction factors for grating-based fluorescence spectrometers in the ultraviolet-visible spectral range. These methods are intended for instruments with a 0/90 transmitting sample geometry. Each method uses different types of transfer standards, including 1) a calibrated light source (CS), 2) a calibrated detector (CD) and a calibrated diffuse reflector (CR), and 3) certified reference materials (CRMs). The wavelength region covered by the different methods ranges from 250 nm to 830 nm with some methods having a broader range than others. Extending these methods to the near infrared (NIR) beyond 830 nm will be discussed briefly, when appropriate. These methods were designed for scanning fluorescence spectrometers with a single channel detector, but can also be used with a multichannel detector, such as a diode array or a CCD.
KeywordsEmission Signal; Fluorescence; Spectral
The title and scope are in draft form and are under development within this ASTM Committee.Back to Top
Paul De Rose
Negative Votes Need Resolution