1.1 This test method covers the measurement of saturation magnetization of magnetic materials using a vibrating-sample magnetometer.
1.2 Explanation of symbols and abbreviated definitions appear in the text of this test method. The official symbols and definitions are listed in Terminology A340.
1.3 The values stated in either customary (absolute (or practical) cgs-emu) units or SI units are to be regarded separately as standard. Within the text, the SI units are shown in brackets. The values stated in each system are not exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in nonconformance with this method.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
A340 Terminology of Symbols and Definitions Relating to Magnetic Testing
induction; magnetometer; magnetic field strength; magnetic test; saturation induction; saturation magnetization; vibrating sample magnetometer;
ICS Number Code 29.030 (Magnetic materials)
ASTM International is a member of CrossRef.
Citing ASTM Standards
[Back to Top]